Lab supply | First online laboratory shop, and selling where new features were introduced. practice. Instrumental factor must be checked. Chema Diagnostica will not be responsible for instrumental bad programming. Mindray BAA. Mindray) owns the intellectual property rights to this Mindray product and this manual. This manual may refer to information protected by copyrights or patents.
|Published (Last):||17 April 2015|
|PDF File Size:||8.76 Mb|
|ePub File Size:||15.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Unique Diagnostics » Semi Automated BA 88A
Click Lower upgr and the Lower computer upgrading dialog box will be displayed. Software List Figure System structure of the software 5. Figure Structure of the CPU extension board 4.
More products from this supplier. When an error or failure occurs, relevant alarm message will be ba88a, and the system will take corresponding actions. Analogue Board Interface 4.
Medicus announces diagnostic-data mobility initiative Mijdraya Vienna-based digital health company, is proud to announce If not, execute the CPU board software upgrading again.
Inappropriate maintenance may lead to unreliable results, or even equipment damage and personal injury. Figure Lamp base assembly structure 3. During configuration, modifying parameters other than temperature parameters should not be allowed. Wash the tubing with distilled water or DI water before shutting down the analyzer for instrument installed with flow cell.
Enter the password—analyzer and click OK. Software Upgrading If the optical surface of the flow cell is contaminated, use cloth soaked with certain amount of absolute alcohol to clean it.
The structure is shown as the following figure. Operational headquarter of our company is situated at Madurai, Tamil Nadu India. The system is an analyzer designed for in vitro quantitative determination of clinical chemistries in serum, plasma, urine and CSF samples.
Semi Auto Analyser in Bengaluru. Pinch the two buckles on the left and right side of the pump shell and detach the pump head carefully.
BAA semi-auto chemical analyzer for clinic. Page 3 This manual may refer to information protected by copyrights or patents and does not convey any license under the patent rights of Mindray, nor the rights of others.
When maintaining the components on the side of analogue board and air duct, it is not necessary to dismount the flow cell base assembly and maintenance can be done on the dismounted photometer assembly. The upgrading time is in direct proportion to the size of the file. Ascom appoints Ljubisav Matejevic to lead the Strategic Alliances business Ascom invests further into its strategic partners business by appointing Do not install devices generating excessive electromagnetic noise around the system.
Before detected by control module, all input signals should go through the level translation circuit. These products are thoroughly examined on several quality parameters before final dispatch.
BA 88A Mindray Semi Auto Biochemistry Analyzer
Page 18 Analogue Board Modification on the screen. Page 72 deviation can not be mundray measured in operating software, so they are configured by other methods. And the following screen will be displayed: Open the optical window cover, use cross screw driver to unscrew the screws on the lamp replacement window cover.
Table of Contents Add to my manuals Add. EndpointFixed-timeKinetics and Absorbance 2. A confirmation has been sent to you, please click the link to verify your email address and activate your subscription. Treating Waste Liquids Please observe the following instructions to prevent environmental pollution and personal injury caused by waste. Tubing should be ba-88w irregularly and replaced in time. Date of Manufacture Manufacturer CE marking.
It has ergonomic interface and in-built thermal printer. Conventions Used in This Manual This manual uses the following typographical conventions to clarify meanings in the text. The photometer assembly is the key for reliability of the system.
If obvious blow-up is found, skewed installation of the lens might be the cause and the readjustment might be needed. Replacing Cpu Board Pull the wire out and you can see the wire joint connecting the lamp assembly, with one end connecting to the lamp and the other end connecting to the inside of the instrument.
Figure Structure of USB functions 4. When disassembling the instrument, do not drag the cable with excessive force.